Technical requirements of the stainless steel orthopedic extension device quipping with all brands operating tables
1. Material and function requirements:
1.1 The main structure of the orthopedic traction frame is required to be made of high-quality 304 stainless steel, suspended design, and equipped with auxiliary trolleys.
1.2. It is required that the surgical table orthopedic extension device is easy to install and disassemble. There is no gap between the traction frame and the table surface after installation, and the stability is high.
1.3. The pelvic frame of the operating table traction frame is required to be connected with the left, middle and right three insertion positions at the end of the operating table to facilitate the left, middle and right movement of the pelvic plate. Arm, C-arm operation has no obstacles, no dead angles, can be completely fluoroscopy, and can enable the operator to operate closer to the patient.
1.4. It is required that the tow bar can change the angle greatly, and the retractable design is used to increase the adjustable range of the tow frame, which is suitable for all people.
2. Technical parameter requirements:
2.1 Telescopic travel of traction frame support arm: ≥540mm
2.2. The first joint's vertical rotation angle of the traction frame: ≥90 °, the second joint's backward rotation angle: ≥180 °
2.3 Rotation angle of ankle anchor: ≥360 °
2.4, traction stroke of tractor: ≥200mm, lifting stroke of tractor: ≥400mm
2.5. Lifting stroke of side body component (side bed): ≥1200mm
2.6 Under the condition of erection of the floor support frame, the traction frame can bear weight: ≥50KG
2.7, trolley height 700mm ± 10mm